The 8407A Rear Panel IF REF output and IF TEST outputs at 278kHz on the the Oscilloscope. This shows the IF is phase-locked to the 8601A.
The 8407A sends 3 outputs to the Display Unit: IF Ref, IF Test Amplitude and IF Test Phase.
Comparative Amplitude Measurements
1. Connect REF & TEST Channels to Power Splitter.
2. Use appropriate DIRECT or ATTN inputs, depending on input level.
3. Terminate other unused inputs.
4. Check REF CHAN LEVEL is in OPERATE window, adjust RF output from Generator if necessary, or use ADJ (normally in centre position) with ±20dB steps if necessary.
5. Use DISPLAY REF and AMPL VERNIER to zero Meter with 3dB range selected.
6. Connect Oscilloscope to AMPLITUDE 50mV/dB output and check 0mV output.
7. Connect Device Under Test (DUT) to TEST Channel.
8. Read out AMPLITUDE dB on Meter for gain or loss, eg. -20dB on 30dB range.
9. Confirm reading on Oscilloscope, eg. -1V.
If the gain or loss is >30dB, then adjust the DISPLAY REFERENCE attenuator to bring the Meter closer to zero. The algebraic result of the Meter and attenuator provides the gain or loss.
Comparative Phase Measurements
1. Set up for Amplitude Measurement as above.
2. Set PHASE OFFSET switch to zero.
3. Select 180° Phase range.
4. Adjust PHASE VERNIER to zero Meter.
5. Select 60°, then 18°, then 6° Phase range, adjusting PHASE VERNIER if required to zero Meter.
6. Connect Oscilloscope to PHASE 10mV/DEG output and check 0mV output.
7. Connect Device Under Test (DUT) to TEST Channel.
8. Read out PHASE DEGREES on Meter for phase shift, eg. -24° on 60° range.
9. Confirm reading on Oscilloscope, eg. -240mV.
In this case a 6dB attenuator is connected to the TEST channel.
The Meter indicates -6.5° and the Oscilloscope shows -65mV.
This is for a CW of 110MHz, the phase varies from -2° to -6.5°
non-linearly over the 100kHz to 110MHz frequency range.
When measuring large phase angles near 180° it is best to use the PHASE OFFSET control to bring the Meter reading closer to zero to avoid instability. The algebraic result of the Meter and control provides the phase angle.
These are measured in the same way except that the 8721A Directional Bridge is inserted in the TEST output of the Power Splitter. SOURCE to Power Splitter, LOAD to DUT, and REFLECTED to TEST CHANNEL of 8407A. Note that the cable lengths in the REF and TEST channels must be matched and that there is 6dB insertion loss between SOURCE to LOAD and also between LOAD to REFLECTED.
Reflection measurement of the 6dB attenuator
using the 8721A and matched cables over
a full 110MHz but slow sweep
Calibration is achieved with a BNC Short connected to the LOAD. This is replaced with the DUT for reflection measurements. The output of the DUT must be terminated in 50Ω. See the Operating & Service Manual for the HP 11652A Transmission/Reflection Kit for more information.
Fast swept measurements are difficult to display correlated against frequency, even using the Generator Sweep Output to the Oscilloscope x-axis. This is where the 8412B Phase Magnitude Display comes in. This requires an additional connection from 8601A Sweep output to 8412B Sweep input.
Top trace is phase at 10°/division and
bottom trace is amplitude at 1.0dB/division
over a full fast 110MHz sweep.
Comparative amplitude and phase measurements are made as for the 8413A, except that the calibration is normally made with the trace adjusted to the centre screen graticule. The change after connecting the DUT can be measured by the vertical displacement. The amplitude can be adjusted from 10dB/Div, down through 2.5dB/Div, 1dB/Div & 0.25dB/Div. Phase can be adjusted from 90°/Div down through 45°/Div, 10°/Div & 1°/Div. Phase offset is also available in 20° steps from 0° to ±180°. This is a bit tricky to set up properly and some practice is required!
The Aux RF output of the 8601A can be used to drive a Frequency Counter and in Manual Sweep Mode the horizontal display can be calibrated. For instance, the frequency of a phase change transition can be measured.
The top trace shows a 1dB amplitude change over 1MHz to 110MHz
The bottom trace shows a 360° phase change at 58MHz
The Operating and Service Manual for the 8412B explains how to make: swept frequency transmission measurements; and swept frequency reflection measurements. The 8412B is intended for swept frequency measurements with a fast sweep speed.
The 8414A Polar Display also displays magnitude and phase information but in polar form.
To calibrate the display follow steps 1 to 4 as for the 8413A then:
5. Press and hold the BEAM CTR push-button and adjust HORIZ POS & VERT POS to position the dot in the centre of the graticule.
6. Without the DUT in circuit and with equal cable lengths to the REF and TEST inputs.
7. Adjust the DISPLAY REFERENCE attenuators, AMPL VERNIER & PHASE VERNIER to position the dot or cluster for a reference indication of r = 1/0°. This is placing the dot at the outer circle on the 0° radial line.
The DUT can now be inserted in the TEST channel. This calibrates the system for a reflection coefficient of 1.0 at the outer circle.
Display of the reflection characteristics of a RF Transformer
If the DUT has an impedance of 50Ω at all frequencies in the range being swept, the display of its complex reflection coefficient is a dot at the centre of the graticule. If not, the display of the complex reflection coefficient is an irregular pattern which represents at each point: a specific frequency; the reflection coefficient magnitude; and the phase angle. The frequency can be determined by a Counter with the 8601A in Manual Sweep Mode or by frequency markers. The magnitude of the reflection coefficient can be read on the concentric circles with zero at the centre, then 0.2/circle with 1.0 at the outer circle. Changing the DISPLAY REFERENCE gain will change the calibration. Adding 14dB gives outer circle calibration of 0.2 and adding 20dB gives 0.1 calibration. The phase angle (between the incident voltage and the reflected voltage at the plane of measurement) can be read directly in degrees by drawing a radiall line between the centre, the point of interest and the outer circle.
This is a complex subject and more information is provided in the Manual, along with polar display of return loss; polar display of transmission measurements; scattering parameters (S-Parameters); polar display of normalised impedance and admittance; and alternative Smith Charts.
The HP 8418A Auxiliary Display Holder
provides the opportunity for 2 Display Units to be viewed at the same time. This is very useful to verify the measurements of the DUT and to see the sometimes complex effect of making adjustments in the DUT circuit.
Complete Network Analyser System
8721A, Power Splitter, Attenuator Control Unit,
8407A with 8412B
& 8418A with 8414A
It would be more convenient to have this System on a mobile trolley or side table, rather than on the bench, as access is required to the rear panels for cabling.
Finally, I would like to acquire the 11652A Transmission/Reflection Kit which contains: the Power Splitter 11652-60009; the 8721A Directional Bridge; 1250-0929 BNC Short; 11652-60001 Precision BNC 50Ω Terminations; 3 BNC Cable Assemblies 11652-0002,3,4; and 2 other BNC Adaptors. Please contact me if you have one for sale.
This will be used in place of the Mini-Circuits Power Splitter.
Typical applications of the Network Analyser System are:
- Amplifier stability by plotting the open-loop phase-gain margin of a feedback amplifier
Adjusting a tunable filter or resonance of a tuned circuit
Complex impedance or admittance of an aerial
Matching cable electrical length
Cable return loss and VSWR
Group delay from non-linear phase variations
Reflection coefficient of a port
Characteristics of adaptors and connectors
There are various tutorials available on the Web describing these measurements.