PROGRESS AND PROBLEMS
A short review of how far I have got with this project and what's holding me up.
Bearing in mind that I'm starting work on Spectrum Analysis from scratch. I know little about this subject. I have a limited amount of time to spend on this hobby, so work proceeds in occasional short bursts. I generally learn something new every time.
Bought 141T, 8555A [10MHz to 18GHz], 8552B in good condition. Huge learning curve in reading downloaded Manuals. Inspected units and powered up ok. System all working.
Bought damaged 8553B [1kHz to 110MHz] RF Section. Downloaded Manual. Completely stripped down, damaged parts replaced (thanks Robert), a few faults fixed and re-assembled.
Bought untested 8556A [20Hz to 300kHz] LF Section. Downloaded Manual.
Bought 2 off 8443A, 1 in good condition and 1 for spares. Downloaded Manual. Inspected good unit and powered up ok.
Spare unit stripped down to assembly level for spares to resell. Attenuators used to make stand-alone attenuator unit.
Bought 8445B Automatic Preselector with inter-connecting cables. Downloaded Manual. Inspected and powered up ok. Works ok with 141T.
No activity on this project for months! Too many other things going on. At last bought a 8444A Tracking Generator with a Manual.
Ordered the special inter-connect cables for the 8443A & 8444A to 141T from GLK Instruments. Occasionally these cables come up for sale on eBay but tend to sell for a similar price. Although expensive, probably better to buy some new ones. Cables arrived quickly, had forgotten about extra €50 for import duty! Very well made and tested ok.
At last, starting work on system testing. Apart from lack of time, I first had to get other basic items of test equipment tested such as Signal Generators, Frequency Counters, Power Meters, and DMM's.
Wanted to use the Spectrum Analyser to test some cables and no display! Having moved the Analyser a few times maybe something had become dis-connected. Need to put it on the Bench and check it out. See Repair HP 141T
and Repair HP 8552B
Bought a 08553-60122 50 Ohm Termination, which is used instead of the special cables when the Tracking Generators are not connected.
Minor repair of 8552B and 8555A plus work on Operating Procedures.
Now have a Spectrum Analyser System with Tracking Generators for: low & audio frequencies 20Hz to 300kHz; video & radio frequencies 1kHz to 110MHz; and VHF/UHF & microwave frequencies 10MHz to 18GHz (tracking to 1.3GHz).
Performance testing has been completed as far as possible. Need to investigate if the remaining tests are necessary, as some specialist test equipment is not available.
Also need to study various application notes and information on spectrum analysis.
Will maybe look out for a 8554B RF Section to complete the set, although its frequency range of 100kHz to 1,250MHz is provided by the other RF Sections.